Toy Sharp Edge Tester
Toy Sharp Edge Tester Model#BND-RL06 Usage This sharp edge tester is a precision instrument of durability, which is mainly used to determine whether there is a Sharpe edge in a toy. Reference Standards 16CFR 1500.49 ASTM F963 4.7 EN71-1998.8.11 Descriptions 1 ． Structure （ Figure 1 ）： （Figure1 ）...
Toy Sharp Edge Tester
This sharp edge tester is a precision instrument of durability, which is mainly used to determine whether there is a Sharpe edge in a toy.
ASTM F963 4.7
A: Handle B: Handle Start Switch C: Panel Start Switch
D: Axle E: Pedal Start Switch F: Fixation Device
Thickness of Test Tape：0.066-0.099mm
Width of Test Tape：No less than 6mm
Cylinder Axis: d=9.53mm
Charger：I/P 12A 50HZ 220V 25W O/P 15V 1.5A
Environmental Requirement：Room Temperature
The tester is portable, put it on a level surface when using.
Preparation for test specimen
Place the specimen in a standard room temperature of 20-25 ° C and a relative humidity of 20-70% for at least 4 hours
1.Make sure it’s plugged in the power before the test, then to check if the tester works normally.
1.1.Turn on the power by 3 ways. Choose one way to turn on the power.
Way 1 hold handle start switch.(figure1 B)
Way 2 hold panel start switch.(figure1 C)
Way 3 hold pedal start switch.(figure1E)
The tester must be power on. Otherwise, it won’t work .
1.2 The edge of the toy and the axis line maintaining a angle of 90 ± 5 ° during the test. Apply a force to axis, when the indicator of level is on , it reaches to a force of 1.35LBS as required
2.Covered layer of PTFE tape at the end of axis. Two ends of tape could be closed but the length of overlap cannot exceed 0.1 inch.
3.Take the edge of the toy to touch the PTFT tape covered in axis and maintain a angle of 90 ± 5 ° until the indicator of level is on. Then choose one way to turn on the power as mentioned above so the axis will run for a circle. Take off PTFT tape and observe. If the length of the laceration exceed 12.7mm, it can conclude that this toy is with a sharp edge. If the length of the laceration is less than 12.7mm, no sharp edge exist.
4.Plug out the power line after the test.
1.keep the instrument clean at any time. rub the cylindrical axis with a cotton cloth to keep its net before and after each experiment.
2.It must be coated with anti-rust oil to prevent the rust after each use.
1.What’s your MOQ?
MOQ for most of our products is 1 unit. If you purchase more products, you will get a better price accordingly.
2.Do you have an OEM/OED service?
Yes! Please contact us for details
3.What’s the lead time?
It depends on the product and the qty.
If it is in stock, 3~7 days will be taken. Normally, 1~2 weeks for most products. 4~7 or more weeks will be needed for complex ones.
Please contact us to confirm.
4.What’s the payment term?
T/T, Western Union, Money gram ,PayPal ,etc .
5.What’s the shipping way?
Via air , sea or express like DHL, UPC, TNT, FedEx and so on.